 sensA.type=f
 sensB.type=t sensB.curve=rhfe4140 sensB.alarm=323
 sensC.type=s sensC.curve=std6
 sensD.type=m sensD.curve=std6
 dev="ILL Furnace (1500 K, typ K)"
 tlimit=1500 resist=1000 maxPower=2000 powfact=2000
 controlmode=1
 prop=3 int=0.3 deriv=0
 lscfg="LINEAR A,2,1.6,3,1,-1.4;INTYPE A,0,1,2,10,12;DISPFLD 4,A,4;DISPLAY:4"

