deviceDesc = orange cryofurnace with 70 mm sample space (with power supply)

defineTemperature tt
device stick_menu "ill3 ill3g"
stick ill3

makenv tt -driver [lsdriver] {
  lsc_sensor tm A -sensor rhfe8119 -sensorname VTI -alarm 550
  lsc_sensor ts C -sensorname sample -alarm 550
  lsc_sensor ts_2 D -sensor code -sensorname sample2
  if {[lsdriver] eq "340_lsc"} {
    lsc_loop set -loop 2 -channel A -linearpower -200 -maxheater 200W -resist 20 -color green
  } else {
    error "can not use ILL3_PS with [lsdriver]"
  }
}
tt set/limit 550
makeCCU4 ill3 nv n2 he

source std_orange.table

