deviceDesc = He3 cryo (Anna Soter)

defineTemperature tt

makenv tt -driver 340_lsc {
  lsc_sensor t1 C -sensor X133320
  lsc_sensor t2 D -sensor X133322
  lsc_sensor sh A -sensor X101105 -sensorname shield
  lsc_sensor hx B -sensor UMEN102 -sensorname heatexch
  lsc_loop set -loop 1 -channel B -maxheater 10mA -resist 120Ohm -color green
}

tt set/prop 500
tt set/integ 1000
tt set/limit 3.5
#tt dblctrl/int2 100

makeCCU4 he3ann he
cc hav 1
cc hea 1

# slow time interval
cc hts 40
# fast time interval
cc htf 10
# he full level
cc hfu 1
# he empty level
cc hem 105
# stop filling
cc hh 95
# measuring current (higher than normal, due to turbulences)
cc hd 100

# 250 l vessel:
cc hem0 720

makenv flow -driver bronkgen -port he3anna-ts:3004 flow 1

GraphAdd flow ln/min

makenv p -driver pfeiffermulti -port he3anna-ts:3003

GraphAdd p mbar p1
GraphAdd p.p2 mbar p2
GraphAdd p.p3 mbar p3
GraphAdd p.p4 mbar p4
GraphAdd p.p5 mbar p5
GraphAdd p.p6 mbar p6
