# Detector settings commands

Commands to setup the settings of the detector

- Settings, trim & cal Directories: commands to setup settings/trim/cal directories
- Settings and Threshold: commands to configure settings and threshold of detector
- DACs: commands to configure DACs of detector
- ADCs: commands to readout ADCs of detector

## Settings, trim & cal Directories

commands to setup settings/trim/cal directories

- settingsdir [dir] Sets/gets the directory where the settings files are located. `Returns` ``(string) dir

- trimdir [dir] obsolete `settingsdir`. `Returns` ``(string) dir

- caldir [dir] Sets/gets the directory where the calibration files are located. `Returns` ``(string) dir

- trimen [n e0 e1...e(n-1)] Sets/gets the number of energies n at which the detector has default trim file and their values in eV (int). `Returns` ``(int int...) n e0 e1...e(n-1)

## Settings and Threshold

commands to configure settings and threshold of detector

- settings [s] sets/gets the settings of the detector. Options: `standard`, `fast`, `highgain`, `dynamicgain`, `lowgain`, `mediumgain`, `veryhighgain`, `lownoise`, `dynamichg0`, `fixgain1`, `fixgain2`, `forceswitchg1`, `forceswitchg2`.
  In Eiger, only sets in client shared memory. Use `threshold` or `thresholdnotb` to pass to detector. Gets from detector. `Returns` ``(string) s

- threshold [eV] [sett] sets/gets the detector threshold in eV. sett is optional and if provided also sets the settings. Use this for Eiger instead of `settings`. `Returns` ``(int)

- thresholdnotb [eV] [sett] sets/gets the detector threshold in eV without loading trimbits. sett is optional and if provided also sets the settings. Use this for Eiger instead of `settings`. `Returns` ``(int)

- trimbits [fname] loads/stores the trimbits to/from the detector. If no extension is specified, the serial number of each module will be attached. `Returns` ``(string) fname

- trim:[mode] [fname] trims the detector according to mode and saves resulting trimbits to file. Mode: noise, beam, improve, fix. Used in MYTHEN only. Only put! `Returns` ``("done")

- trimval [i] sets all trimbits to i. Used in EIGER only. `Returns` ``(int)

- pedestal [i] starts acquisition for i frames, calculates pedestal and writes back to fpga. Used in GOTTHARD only. Only put! `Returns` ``(int)

## DACs

commands to configure DACs of detector

- vthreshold [i] [mv] Sets/gets detector threshold voltage for single photon counters. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- vcalibration [i] [mv] Sets/gets the voltage of the calibration pulses. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- vtrimbit [i] [mv] Sets/gets the voltage to set the width of the trimbits. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- vpreamp [i] [mv] Sets/gets the voltage to define the preamplifier feedback resistance. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- vhaper1 [i] [mv] Sets/gets the voltage to define the feedback resistance of the first shaper. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- vshaper2 [i] [mv] Sets/gets the voltage to define the feedback resistance of the second shaper. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- vhighvoltage [i] Sets/gets the high voltage to the sensor in V. `Returns` ``(int ["mV"]).

- vapower [i] Sets/gets the analog power supply for the old chiptest board in DAC units. `Returns` ``(int ["mV"])

- vddpower [i] Sets/gets the digital power supply for the old chiptest board in DAC units. `Returns` ``(int ["mV"])

- vshpower [i] Sets/gets the comparator power supply for the old chiptest board in DAC units. `Returns` ``(int ["mV"])

- viopower [i] Sets/gets the power supply of the FPGA I/Os for the old chiptest board in DAC units. `Returns` ``(int ["mV"])

- vrefds [i] [mv] Sets/gets vrefds. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- vcascn_pb [i] [mv] Sets/gets vcascn_pb. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- vcasc_pb [i] [mv] Sets/gets vcasc_pb. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- vout_cm [i] [mv] Sets/gets vout_cm. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- vcasc_out [i] [mv] Sets/gets vcasc_out. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- vin_com [i] [mv] Sets/gets vin_com. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- vref_comp [i] [mv] Sets/gets vref_comp. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- ib_test_c [i] [mv] Sets/gets ib_test_c. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- dac[0..7] [i] [mv] Sets/gets dac[0..7] for MOENCH02. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- vsvp [i] [mv] Sets/gets vsvp. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- vsvn [i] [mv] Sets/gets vsvn. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- vtr [i] [mv] Sets/gets vtr. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- vrf [i] [mv] Sets/gets vrf. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- vrs [i] [mv] Sets/gets vrs. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- vtgstv [i] [mv] Sets/gets vtgstv. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- vcmp_ll [i] [mv] Sets/gets vcmp_ll. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- vcmp_lr [i] [mv] Sets/gets vcmp_lr. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- vcal_l [i] [mv] Sets/gets vcal_l. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- vcomp_rl [i] [mv] Sets/gets vcomp_rl. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- vcomp_rr [i] [mv] Sets/gets vcomp_rr. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- rxb_rb [i] [mv] Sets/gets rxb_rb. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- rxb_lb [i] [mv] Sets/gets rxb_lb. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- vcp [i] [mv] Sets/gets vcp. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- vcn [i] [mv] Sets/gets vcn. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- vis [i] [mv] Sets/gets vis. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- iodelay [i] [mv] Sets/gets iodelay. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- dac:j [i] [mv] Sets/gets value for DAC number j for the new chiptestboard. Normally in DAC units unless `mv` is specified at the end of the command line. `Returns` ``(int ["mV"])

- adcvpp [i] Sets/gets the Vpp of the ADC 0 -> 1V ; 1 -> 1.14V ; 2 -> 1.33V ; 3 -> 1.6V ; 4 -> 2V . `Returns` ``(int ["mV"])

- v_a [i] mv Sets/gets value for Va on the new chiptest board. Must be in mV. `Returns` ``(int ["mV"])

- v_b [i] mv Sets/gets value for Vb on the new chiptest board. Must be in mV. `Returns` ``(int ["mV"])

- v_c [i] mv Sets/gets value for Vc on the new chiptest board. Must be in mV. `Returns` ``(int ["mV"])

- v_d [i] mv Sets/gets value for Vd on the new chiptest board. Must be in mV. `Returns` ``(int ["mV"])

- v_io [i] mv Sets/gets value for Vio on the new chiptest board. Must be in mV. `Returns` ``(int ["mV"])

- v_chip [i] mv Sets/gets value for Vchip on the new chiptest board. Must be in mV. `Returns` ``(int ["mV"]). Normally don't use it!

- v_limit [i] mv Sets/gets a soft limit for the power supplies and the DACs on the new chiptest board. Must be in mV. `Returns` ``(int ["mV"])

## ADCs

commands to readout ADCs of detector

- temp_adc Gets the ADC temperature. `Returns` `EIGER`,JUNGFRAU(double"°C") Others ``(int"°C")

- temp_fpga Gets the FPGA temperature. `Returns` `EIGER`,JUNGFRAU(double"°C") Others ``(int"°C")

- temp_fpgaext Gets the external FPGA temperature. Used in EIGER only. `Returns` `EIGER`(double"°C")

- temp_10ge Gets the 10Gbe temperature. Used in EIGER only. `Returns` `EIGER`(double"°C")

- temp_dcdc Gets the temperature of the DC/DC converter. Used in EIGER only. `Returns` `EIGER`(double"°C")

- temp_sodl Gets the temperature of the left so-dimm memory . Used in EIGER only. `Returns` `EIGER`(double"°C")

- temp_sodr Gets the temperature of the right so-dimm memory. Used in EIGER only. `Returns` `EIGER`(double"°C")

- adc:j Gets the values of the slow ADC number j for the new chiptest board. `Returns` ``(int"°C")

- temp_fpgal Gets the temperature of the left frontend FPGA. Used in EIGER only. `Returns` `EIGER`(double"°C")

- temp_fpgar Gets the temperature of the right frontend FPGA. Used in EIGER only. `Returns` `EIGER`(double"°C")

- i_a Gets the current of the power supply a on the new chiptest board. `Returns` ``(int"mV")

- i_b Gets the current of the power supply b on the new chiptest board `Returns` ``(int"mV")

- i_c Gets the current of the power supply c on the new chiptest board `Returns` ``(int"mV")

- i_d Gets the current of the power supply d on the new chiptest board `Returns` ``(int"mV")

- i_io Gets the current of the power supply io on the new chiptest board `Returns` ``(int"mV")

- vm_a Gets the measured voltage of the power supply a on the new chiptest board `Returns` ``(int"mV")

- vm_b Gets the measured voltage of the power supply b on the new chiptest board `Returns` ``(int"mV")

- vm_c Gets the measured voltage of the power supply c on the new chiptest board `Returns` ``(int"mV")

- vm_d Gets the measured voltage of the power supply d on the new chiptest board `Returns` ``(int"mV")

- vm_io Gets the measured voltage of the power supply io on the new chiptest board `Returns` ``(int"mV")