# Get values from data/<station>/<year>/config_acsm_<year>.r, values used in Tofware analysis.
default_params: &default_params
  IE_tofware: 183.1
  ABRefWave_tofware: 174139
  RIE_SO4_tofware: 0.88
  RIE_NH4_tofware: 3.183
  RIE_Org_tofware: 1.4
  RIE_NO3_tofware: 1.05
  RIE_Chl_tofware: 1.0
  CE_tofware: 1.0
  flow_ref_tofware: 2.21



# Get values from data/<station>/<year>/cal.csv
calibration_intervals: # Specify datetimes (YYYY-MM-DD HH-MM-SS)
  interval_1:
    start_datetime : '2024-01-01 00-00-00' # date and time where calibration was concluded 
    end_datetime : '2024-12-09 14-00-00'
   
    IE_correct: 238.85
    AB_ref_correct: 117359
    RIE_SO4_correct: 0.915
    RIE_NH4_correct: 2.895
    RIE_Org_correct: 1.4
    RIE_NO3_correct: 1.05
    RIE_Chl_correct: 1.0
    CE_correct: 1.0
    flow_ref_correct: 2.315
    << : *default_params

  interval_2 : 
    start_datetime : '2024-12-09 14-00-01'
    end_datetime : '2024-12-31 23-59-59' 
         
    IE_correct: 206.3
    AB_ref_correct: 347000
    RIE_SO4_correct: 0.81
    RIE_NH4_correct: 3.76
    RIE_Org_correct: 1.4
    RIE_NO3_correct: 1.05
    RIE_Chl_correct: 1.0
    CE_correct: 1.0
    flow_ref_correct: 2.24
    << : *default_params

# Idenfity diagnostics and species variables for flag generation. Names must correspond to table column names the corresponding data file  
variables:
  diagnostics:  ['FilamentEmission_mA','VaporizerTemp_C','FlowRate_ccs','ABsamp']
  species: ['Chl_11000','NH4_11000','SO4_11000','NO3_11000','Org_11000']